Ningbo Goldshine Electronic Co., Ltd weighbridge is a kind of scale which is usually composed of three parts: scale body, sensor and instrument. The fixed base has a load-bearing tray or metal plate. So how should the Ningbo Goldshine Electronic Co., Ltd weighbridge be used correctly in daily application? What should be noted after use? Let’s get to know about it together!
1: When using Ningbo Goldshine Electronic Co., Ltd weighbridge, the scale body shall be placed as lightly as possible, shall not be violently placed. After use, each part need to be removed to avoid the damage of the blade.
2: If the mechanical scale is used continuously, check whether the empty scale is balanced after a certain number of times, so as to ensure the accuracy of measurement.
3: In the process of using mechanical balance, besides proofreading empty scale, nothing can be hung on the weight, otherwise the accuracy of weighing will be affected.
4: In order to avoid the damage of the scale body, it is necessary to pay attention to the maximum weight of the object, which shall not exceed the weighing value of the mechanical scale.
Precautions before use:
When using, first, remove the inner and outer packaging of Ningbo Goldshine Electronic Co., Ltd weighing scale and put it on a flat and hard ground, and four wheels are grounded simultaneously to prevent the body from tilting and affecting the measurement.
When using the scale, we should shake the bearing plate slightly first to make sure every part well connected. Meanwhile, check whether the measuring lever is deformed. Then hang the weight and measure the lever force point ring. If it is unbalanced, rotate the adjusting screw to reach the balance of the empty scale.
Maintenance of Ningbo Goldshine Electronic Co., Ltd weighbridge:
1: After use for a long time, the scale such as the blade and the bearing ring, etc., which will be worn and tired, and the accuracy and sensitivity will be reduced mutually. Therefore, the Ningbo Saintbond weighbridge must be checked and calibrated regularly. In general, it shall be overhauled once a year or half a year in general, at least once a year.
2: The calibrator shall be kept dry and clean frequently, and the liquid substance and dust shall not be left in the sight frame and all the tool supports. The hook of the scale shall be kept clean frequently to avoid affecting the sensitivity.
3: To keep it intact, do not remove it at will.
4: During the handling, the impact shall be strictly prevented and the measuring lever shall not be lifted by the bar, otherwise the scale body and components will be damaged, and the measuring lever will be deformed, which will affect the accuracy of Ningbo Goldshine Electronic Co., Ltd weighing scale.
5: Do not put the scale in the open air, do not put the scale with corrosive items to avoid the corrosion components affecting the service life.